The Quick-Control Oscilloscope
can use the following properties.
|Read-only property that provides available channel names in a cell array.|
|Read-only property that provides currently enabled channel names in a cell array.|
|Read-only property that indicates the communication status.|
|Use to get or set a timeout value.|
Value cannot be negative
number. Default is
|Use to get or set acquisition time value. Used to control the
time in seconds that corresponds to the record length.|
Value must be a positive, finite number.
|Use to set or get the length of time in seconds from the trigger
event to first point in waveform record.|
If positive, the first point in the waveform occurs after the trigger. If negative, the first point in the waveform occurs before the trigger.
|Use to set the triggering behavior. Values are:|
|Use to set or get trigger slope value. |
Valid values are
|Specifies the voltage threshold in volts for the trigger control.|
|Specifies the source the oscilloscope monitors for a trigger. It can be channel name or other values.|
|Set up before connecting to instrument. Set with value of your
instrument’s resource string, for example:|
set(myScope, 'Resource', 'TCPIP0::a-m6104a-004598::inst0::INSTR');
|Optionally used to set up criteria for connection. |
|Use only if set |